Figure 2-12. ABS Inflation
Check air beam manifold (1, figure 2-13) and ensure that shut-off valves (2) on flexible hoses (3)
connected to each air beam (4) are completely open (in-line with hoses).
Outer TALP and ambulatory airlock doors are stowed on HMT.
Inner ambulatory and TALP airlock doors are stowed in LMS.
TALP rails, recirculation filters and medical chests are stowed in LMS.
Locate inner TALP airlock door (1, figure 2-14) and TALP rails (2) in LMS.
Under non-NBC conditions, inner and outer ambulatory airlock and TALP airlock doors can
be installed at the same time.
As door is worked into opening, avoid pressing fastener hook and fastener loop tape in place.
Ambulatory airlock doors open inward.
Attach TALP fabric tunnel (3) from bottom up to inner TALP airlock door (1) by placing fabric
tunnel (3) between inner TALP airlock door (1) and retaining clamp (4), making sure grommets
engage airlock door (1) posts. Engage latch clamp (5) to secure fabric tunnel (3) to inner TALP
airlock door (1).